Sperm counting silde for Spermolyzer/sperm slide with ruby
Place of Origin:Jiangsu, China (Mainland)
1.Sperm Counting Chamber
2.used in the Fluid samples of the routine examination and measurement inspection,eg. semen, prostate
Macro Sperm Counting Chamber
Sperm counting chamber is widely used in the Fluid samples of the routine examination and measurement inspection,
such as semen, prostate, cervical mucus and so on.
2.Main technical parameters
1) exterior dimensions:76×36×7
2) The grid sperm count plate:0.01mm2/ per Trellis
3) with & without grid sperm counts board
4) The space between pedestal of count plate and cover plate is 0.01mm(10u)
5) precision error: The clearance at 37 degrees is ±0.001 (4 red light wavelengths)
1) Φ 6.5 mm round little platform on the base is carrier liquid platform.
2) Disinfecting the platform using medicine alcohol
3) Dropping the inspected body fluid on theround platform
4) Close the cover and insert it to the microscope platform of constant temperature. Then observe it using 20 X
5) In the microscope visual system, connect digital camera with computer, and then the computer scanning count
work ( still using 20 X optical lens).
1) This product is a check try products with high precision glass and cannot be hitted.
2) Do not use hard thing to blow the surface of the platform.
3) It should be cleaned up without using and put it in special box after disinfection
4) Disinfection method:clean up with clear water first then wipe with medicine alcohol , notice that water
temperature can not high than 40 °C.Do not use high temperature to disinfect.
5.The range of repairing
1) Count board: 1 year (according to the type of bord)
2) The gem fall off (but not lost)
3) The clearance is too high and the normal range is 0.01 + /-0.002. (Can not be judged by visual inspection)
4) The piece need to be fixed , when it come back to the factory, the type of the bord should be accordance
with the type in certificate.After repairing, return it as the former type.
6.The following circumstances are not in guaranteed repair range:
1) The counting plate belongs to senior glass product, vulnerable so please operate it according to instruments.
2) It appears scratch, broken edge glass, the loss of the stones because of improper use.
3) The gem fall off due to the temperature deviation(more than 50 °C, below -4 °C)
4) It appears clearance after using 1 year.
We are Nailfold Capillary Microscopy | Capillaroscopy| Live blood Analysis Microscope| Veterinary Instrument| Andriatrics &Urology Health Care| Lab Equipment| Ultrasonic Diagnostic System| Colposcope Digital Imaging system| Patient Monitoring System| Medical Equipment| Therapeutic| Physiotherapeutic instrument| ECG| ENT Units| Gynecological Device| Infrared Mammary Diagnostic Instrument| LED X Ray Film Viewer| Medical headlamp| Endoscope &Light source,manufacturers in China.Unified Wholesale price.Welcome to inquiry and OEM.